It could remove hardened set MTA easily and securely.
• 0.5mm rigid ball tip
remove MTA that is set at middle 1/3
• 0.3mm bendable tip
remove MTA that is set at apical 1/3
• 0.25mm Controlled Memory tip
remove MTA that is set at apical 1/3 or foreign substance in canal.
• Set scaler with the lowst output state.
• Remove by contacting gently under enough wet condition. Do not press too much.